Some people like to draw an analogy to water to describe voltage. A water pump is like a voltage supply (also known as a battery).
The pump pushes water through a hydraulic system, and the voltage supply pushes electrons through an electronic system.
The higher the rated pressure of the pump, the more 'work' the water can do.
Likewise, the higher the voltage the more 'work' (Watts) the electrons can do.
Voltage is used to provide power (via a battery or wall plug) and its also used as a way of transmitting data. For example, music is recorded from a microphone as an analog voltage signal, if that voltage waveform is applied to a speaker the voltage performs the work of making air move and produces sound.
Voltage is also used to in digital circuits to talk back and forth in binary, usually 5V or 3.3V is a "1" and 0V is a "0", by alternating the 1's and 0's millions of times a second, data can be moved around rather quickly.
Not just an 80's hair metal band! Voltage comes in two flavors (yum): Alternating Current (AC) and Direct Current (DC). Here is a quick tour of the differences.
Direct current voltage is what comes out of batteries. The battery is at 9V, and it pretty much keeps that voltage constant, until it dies. The chemical reactions inside the battery creates DC voltage.
Electronic circuits really like DC voltage.
Alternating current voltage is what comes out of the wall. We call it 120 VAC (Volts Alternating Current) because the generator at the US power plant creates a voltage that oscillates (alternates). At the outlet in your home, the voltage is not constant but goes from about -120V to 0 to +120V to 0 again, 60 times a second. In Europe, it's referred to as 240VAC because the voltage goes from about -240V to +240V at 50 times a second.
Technically the voltage is truly +170V to -170V in the US, which would make 120VAC we're mentioning -> 340Volts peak to peak. Since multimeters tend to show RMS voltages, its easier to just refer to it as 120VAC and remember that the peak postive and negative voltage are each ~1.5x the RMS voltage and the peak-to-peak is going to be ~3x as what the multimeter is displaying for sinusoidal/wall outlet waveforms! You can read all about peak to peak vs RMS voltages here
AC voltage is great for power plants because its easy to transform AC voltages (using a transformer ) up to 50KVAC for long distance travel and then down to 240VAC or 120VAC to safely power your home. Those big honking grey things that you see next to buildings that hum are the huge transformers.
Motors (like your washing machine and refrigerator compressor pump) also like running off of AC voltage.
You can turn AC voltage into DC voltage very easily by using a very small transformer to bring the 120VAC down to a reasonable level like say 16VAC and then rectify it. This is basically what's inside a wall wart plug or your laptop power supply.
Its much harder to turn DC into AC, you will need an inverter which are more expensive than transformers/rectifiers.
Batteries only supply DC voltage and wall plugs only supply AC voltage. However, it is totally possible to have both AC and DC voltage at a certain point:
If an AC voltage is oscillating between -60V and +60V it has 120Vpp AC and 0V DC because the average voltage of -60V and +60V is 0V.
If an AC voltage is oscilating between 0V and 120V then it has 120Vpp AC and 60V DC because the average voltage of 0V and 120V is 60V.
- Test if your power supply is working, are you getting 5V out of that 7805 regulator?
- Verify that your circuit is getting enough power: when all of the blinky lights are on, is the power supply drooping too low?
- Verify signals to and from chips to make sure they are what you expect once the circuit is up and running
- Testing batteries, solar cells, wall plugs, and power outlets (carefully!)
- With a current sense resistor you can perform current testing on a project without possibly damaging your meter.
Voltage is always measured between two points There is no way to measure voltage with only one probe, it is like trying to check continuity with only one probe. You must have two probes in the circuit. If you are told to test at a point or read the voltage at this or that location what it really means is that you should put the negative (reference, ground, black) probe at ground (which you must determine by a schematic or somewhere else in the instructions) and the positive (red) probe at the point you would like to measure.
If you're getting odd readings, use a reference voltage (even a 9V battery is a reasonable one) to check your voltage readings. Old meter batteries and wonky meters are the bane of your existence but they will eventually strike! Good places to take reference voltages are regulated wall plugs such as those for cell phones. Two meters might also be good :)
Voltage is directional If you measure a battery with the red/positive probe on the black/negative contact and the black probe on the positive contact you will read a negative voltage. If you are reading a negative voltage in your ciruit and you're nearly positive (ha!) that this cannot be, then make sure you are putting the black probe on the reference voltage (usually ground)
DC voltage and AC voltage are very different Make sure you are testing the right kind of voltage. This may require pressing a mode button or changing the dial.
Unless otherwise indicated, assume DC voltages
Multimeters have different input impedences that affect readings of high impedence circuits For example, measuring a sensor that has 1Mohm impedence with a 1Mohm impedence meter will give you only half the correct reading
This ranged meter has 5 ranges, the top range is 750 VAC or 1000 VDC, to switch between DC and AC you need to press the DC/AC button on the upper right.
When the probes are not connected to anything, they should display 0V. They might flicker a bit if they pick up ambient voltage (your home is a big radiator of 60Hz voltage which can couple into your meter probes).
The first battery we'll test is a new 1.5V alkaline. This one is a AAA but a AA, C or D cell will be the same voltage. Set the range to 2V DC.
We read 1.588V, which you may think is a mistake, after all its a 1.5V battery so shouldn't it be 1.5V? Not quite, the 1.5V written on the side is just a nominal voltage, or the "average" you may expect from the battery.In reality, an alkaline battery starts out higher, and then slowly drifts down to 1.3V and then finally to 1.0V and even lower. Check out this graph from Duracell's page about alkaline battery voltage.
Using this graph you can easy tell how fresh your battery is and how long you can expect it to last.
Next, we measure a 9V alkaline battery. If we still have the range set to 2VDC we will get a mysterious "1. " display, indicating is it over-range.
For this new battery we get 9.6V. Remember that battery voltage is nominal, which means that the "9V" is just the average voltage of the battery. In reality, it starts out as high as 9.5V and then drops down to 9 and then slowly drifts to 7V. You can check out the discharge curve in the Duracell 9V datasheet
If we want to check a rechargeable AA battery, and it's set to a 20VDC range, we will read 1.3V, which is about what a fully charged NiMH battery will measure.
The first kind we will test is a transformer-based adapter.
Yow! 14V? That's not anything like the 9V on the package, is this a broken wall wart? Turns out, its totally normal. Transformer-based wall adaptors are (almost always) unregulated, which means that the output is not guaranteed to be a particular value, only that it will be at least what is printed on the box. For example, with this adapter it means that when drawing 300mA, the voltage is guaranteed to be higher than 9V.
Since the output is unregulated, the voltage supplied will droop as more current is pulled from it, which means that open-circuit (connected to nothing) the measured output can be as high as 14V. Our power supply tutorial on transformer-based wall adapters covers this in detail
Next, lets check out a Switch-mode adapter.
Switch-mode wall adapters are regulated which means that the output doesn't droop from open-circuit to full load. Its not an ultra-high quality supply, the voltage is 12.2V which is less than 5% error. Still, its much better than the transformer's 50% error!
Lastly, we'll test a 9VAC adaptor, which outputs AC voltage instead of DC. Basically this means that there's still a transformer inside, but no rectifier. This is also an unregulated supply.
Note again that the label says transformer. It requires 120VAC input, US power only. The nominal output is 9VAC at 300mA. The output is indicated twice, once at the top "AC/AC" and then again in the output designator "9V AC"
There is no polarity because AC adaptors are not polarized: AC power oscillates between positive and negative voltages.
We test the output, but get 0V! That's when we remember that the multimeter has to be in AC voltage mode.
For example, here is a few attempts to measure the VCO output of a x0xb0x as seen in the oscilloscope output shown here (its the same one from above).
We read 6.75V DC, which is about right.
However, when trying to measure AC, this multimeter gives us a seemingly random number. (Maybe the DC voltage * 2 ?).